Where to place your Communicator:
- Near a telephone wall jack and electrical outlet that are easily accessible
- Close to where you sleep or near your bedside, within 10 feet (3 meters). If this isn’t possible, place your Communicator where you spend a considerable amount of time each day
- Where you can sit comfortably and see the front of the Communicator
- Where the Communicator and all its cables and accessories will be kept dry and not be exposed to humidity or potential water contact
Set up instructions
Be sure to stay close to the Communicator while completing the steps below to ensure the best connection between your implanted device and the Communicator.
1. Attach the power cord.
Insert the power cord (included) into the jack as shown below and plug the cord into an electrical outlet that is easily accessible.
- The LATITUDE™ indicator will flash yellow for up to one minute.
- All the Communicator indicators will light for approximately one second.
- If the LATITUDE indicator is not lit, check that both ends of the power cord are plugged in firmly. Check if the light on the power cord is lit.
2. Plug cord into the jack.
Plug one end of the Communicator telephone cord (included) into the jack as shown below and the other end of the cord into the telephone jack on the wall.
Note: If you have DSL internet service, you may need to use a DSL filter between the telephone wall jack and the Communicator. Refer to "DSL Internet Service" on page 71 of the LATITUDE Patient Manual for more information.
Optional: To use a telephone with this wall jack, you may plug your telephone into the jack on the Communicator indicated by phone icon. Your Communicator and a telephone can share the same telephone wall jack. However, they cannot be used at the same time.
3. Press the heart button.
4. Wait while the data sends.
Your Communicator’s wave lights will flash green in sequence and repeat for several minutes. If you have previously completed the initial setup, the “Heart Button” will not flash at this point.
Collecting data from your implanted device
If the wave lights are lit a solid green as shown below, your Communicator has successfully connected to the LATITUDE system. The lights will turn off automatically after two minutes.
Troubleshooting your device
If this process takes longer than several minutes, software download and installation may be occurring. Refer to the Software Download and Installation on page 35 of the LATITUDE Patient Manual to troubleshoot.
If the wave lights are not lit a solid green, refer to Troubleshooting on page 46 of the LATITUDE Patient Manual.