Indications, Safety and Warnings
LATITUDE™ NXT Patient Management System
The LATITUDE™ NXT Patient Management System is intended for use to remotely communicate with a compatible pulse generator from Boston Scientific CRM and transfer data to a central database. The LATITUDE NXT System provides patient data that can be used as part of the clinical evaluation of the patient.
The LATITUDE NXT Patient Management System is contraindicated for use with any implanted device other than a compatible Boston Scientific implanted device. Not all Boston Scientific implanted devices are compatible with the LATITUDE NXT System. For contraindications for use related to the implanted device, refer to the System Guide for the Boston Scientific implanted device being interrogated.
Alerts may appear on the LATITUDE NXT website on a daily basis. Primary notification of alert conditions is through the View Patient List page on the LATITUDE NXT website. The clinician needs to log onto the LATITUDE NXT website in order to receive alerts. Although secondary notification through email and SMS
text messages is available, these reminders are dependent on external systems and may be delayed or not occur. The secondary notification feature does not eliminate or reduce the need to check the website. Implanted device data and alerts are typically available for review on the LATITUDE NXT website within 15 minutes of a successful interrogation. However, data uploads may take significantly longer (up to 14 days). If the Communicator is unable to interrogate the implanted device or if the Communicator is unable to contact the LATITUDE NXT server to upload data, up to 14 days may elapse before the LATITUDE NXT server detects these conditions and informs the clinic user that monitoring is not occurring. If both of these conditions occur at the same time, this notification could take up to 28 days. Implanted device data and alert notification may be delayed or not occur at all under various conditions, which include but are not limited to the following: System limitations; the Communicator is unplugged; the Communicator is not able to connect to the LATITUDE NXT server through the configured phone system; the implanted device and the Communicator cannot establish and complete a telemetry session; the Communicator is damaged or malfunctions; the patient is not compliant with prescribed use or is not using the
LATITUDE NXT System as described in the patient manual; if subscribed to the LATITUDE Cellular Data Plan, missing two or more payments discontinues the subscription; the clinic user can identify any patients that are not being monitored as described above by using the Not Monitored filter on the View Patient List.
The LATITUDE NXT System does not provide continuous real-time monitoring. As a remote monitoring system, the LATITUDE NXT System provides periodic patient monitoring based on clinician configured settings. There are many internal and external factors that can hinder, delay, or prevent acquisition and delivery of implanted device, sensor, and patient information as intended by the clinician. These factors include: implanted device clock; patient environment; cellular data service; telephone system; communicator memory capacity; clinic environment; schedule/configuration changes; or data processing.
Refer to the product labeling for specific instructions for use. Rx only. (Rev. D) 046774 AH
LATITUDE™ Patient Management System
The LATITUDE™ Patient Management system is intended for use to remotely communicate with a compatible Boston Scientific pulse generator (PG) and transfer data to a central database. The LATITUDE System provides patient data than can be used as part of the clinical evaluation of the patient.
The LATITUDE System is contraindicated for use with any PG other than a compatible Boston Scientific PG. Not all Boston Scientific PGs are compatible with the LATITUDE System. For contraindications for use related to the PG, refer to the System Guide for the PG being interrogated.
The LATITUDE System is designed to notify clinicians within 24 hours if new PG red alert conditions are detected by the Communicator. Alert notifications are based on clinician configured alert settings. PG data is typically available for review on the LATITUDE System within 15 minutes of a successful interrogation. However, data availability and red alert notification can take up to 24 hours or the next business day. Note that PG data will not be available and alert notification cannot occur if:
- The Communicator is unplugged.
- The Communicator is not able to connect to the LATITUDE System through an active phone line.
- The PG and the Communicator cannot complete a telemetry session. This session must be initiated by the patient if he or she has a pulse generator that uses inductive telemetry.
- The Communicator is damaged or malfunctions.
- The patient is not compliant with prescribed use or is not using the LATITUDE System as described in the patient manual.
Up to two weeks may elapse before LATITUDE System first detects the conditions mentioned above. Additional time may be required for clinic notification and resolution of the condition. During this time, no new patient data, device data, or alert notifications since the last successful data transmission are available.
Refer to the product labeling for specific instructions for use. Rx only. (Rev O) 046774 AH