FFR guided PCI with new COMET™ FFR Pressure guidewire reveals the true picture
This case highlights:
- New Asahi co-developed COMET™ FFR Pressure guidewire’s ability to assess and perform PCI with multiple devices and reconnections
- How FFR guidance changes treatment strategy & alerts the operator to the need for further intervention in sub optimal PCI – where ischemia is still present
- 70 year old man with previous PCI to RCA
- Baseline angiography showed diffuse disease and calcium of the mid LAD
- Pre dilation
- FFR pre and post treatment after each procedural step
COMET™ FFR Pressure Guidewire Characteristics
- Asahi co-developed for true workhorse capabilities
- Free rotation while steering and reliable re-connection
- Optical pressure sensor
- Easy to use IVUS / FFR multimodality interface
Clarify treatment strategy with FFR
- FFR with the new COMETTM Pressure guidewire (0.14’’ Boston Scientific / Asahi Intecc) was selected to interrogate the lesion.
- COMET™ crossed the lesion easily and resting FFR showed a value of 0.76 (borderline value). Intracoronary adenosine was given and the value dropped to 0.59 (Figure 2)
Pre Dilatation Reveals Calcium
- The COMET™ Pressure guide wire was disconnected and the lesion was pre dilated with a 2.5 x 20mm EMERGE semi compliant balloon. Balloon inflation revealed significant calcium
- Due to significant calcium, the vessel was then treated with a non-compliant 2.5 x 15mm NC Quantum Apex. Next, the COMET™ pressure guidewire was re-connected and the resting Pd/Pa was 0.80. (not pictured)
- Next, a Promus Premier 2.5 x38 DES was implanted (Figure 4) and inflated (Figure 5) and post dilated with a non-compliant balloon 2.5x20 mm
Post Stenting FFR performed, edge dissection discovered
- COMET™ was reconnected and IC adenosine was administered, giving an FFR value of 0.73. After further analysis an edge dissection was discovered at the distal stent edge.
- A Promus Premier 2.5 x 8mm DES was implanted to cover the distal edge dissection. After stent implantation and post dilation with NC Quantum Apex FFR showed yet another FFR value under 0.80.
Post Stenting FFR performed, plaque shift discovered
- Further analysis showed that the limited blood flow was caused by plaque shift due to the initial ballooning with long inflation times. To finish the case, the operator decided to implant the final 2.5 x12mm Promus Premier DES distal to the previous stents (Figure 7) with an FFR value of 0.84
Post Stenting FFR performed, non ischemic result
- In this procedure, the use of FFR helped to clarify the continued presence of ischemia, requiring additional treatment methods.
- It’s important to note that the last FFR value was negative for ischemia, demonstrating the efficacy of FFR to guide the entire procedure, achieving an optimal final result.
- Numerous disconnections and reconnections of the new COMET™ FFR Pressure guidewire helped the operators to start, treat and perform final measurements without making the procedure longer and more expensive.
Join Our Complex PCI Community
Stay up-to-date via email on the latest advances in complex PCI.Sign-up Today
Education & Training
Educare offers a comprehensive suite of education and training programs for Complex PCI.Learn More
Download Case Study