For health care professionals in EUROPE excepted those practicing in France as the following pages are intended to all International health care professionals and are not in compliance with the French Advertising law N°2011-2012 dated 29th December 2011 article 34. Other health care professionals should select their country in the top right corner of the website.
Please note that the following pages are exclusively reserved for health care professionals in countries with applicable health authority product registrations. To the extent this site contains information, reference guides and databases intended for use by licensed medical professionals, such materials are not intended to offer professional medical advice. Prior to use, please consult device labeling for prescriptive information and operating instructions.
This Website is protected by the laws on copyright and by the relevant international conventions. It is strictly forbidden to make copies, whether partial or total and on whichever media without prior approval.
The 2D Helical shape is designed to provide optimal anchorability, as the helical shape allows the coil to be in constant, consistent contact with the wall of straight or tapered vessel.
The Diamond shape is designed to maximize occlusive power by tapering the distal and proximal ends of the coil, maximizing cross-sectional flow disruption.
The Interlock - 18 Fibered IDC Occlusion System is designed to be delivered through a 0.021 in inner lumen microcatheter [0.53 mm]
Single Marker Microcatheter
Double Marker Microcatheter
|M001361480||Interlock - 18 Coil||2||40||2D|
|M001361490||Interlock - 18 Coil||2||60||2D|
|M001361500||Interlock - 18 Coil||3||60||2D|
|M001361510||Interlock - 18 Coil||3||120||2D|
|M001361520||Interlock - 18 Coil||4||80||2D|
|M001361530||Interlock - 18 Coil||4||150||2D|
|M001361540||Interlock - 18 Coil||5||80||2D|
|M001361550||Interlock - 18 Coil||5||150||2D|
|M001361560||Interlock - 18 Coil||6||100||2D|
|M001361570||Interlock - 18 Coil||6||200||2D|
|M001361580||Interlock - 18 Coil||8||200||2D|
|M001361590||Interlock - 18 Coil||10||200||2D|
|M001361600||Interlock - 18 Coil||10||300||2D|
|M001361610||Interlock - 18 Coil||12||200||2D|
|M001361620||Interlock - 18 Coil||12||300||2D|
|M001361630||Interlock - 18 Coil||14||200||2D|
|M001361640||Interlock - 18 Coil||14||300||2D|
|M001361920||Interlock - 18 Coil||10||500||2D|
|M001361930||Interlock - 18 Coil||14||500||2D|
|M001361940||Interlock - 18 Coil||18||500||2D|
|M001361950||Interlock - 18 Coil||20||500||2D|
|M001361960||Interlock - 18 Coil||22||600||2D|
|M001361740||Interlock - 18 Coil||2/3||230||Diamond|
|M001361750||Interlock - 18 Coil||2/4||410||Diamond|
|M001361760||Interlock - 18 Coil||2/5||580||Diamond|
|M001361770||Interlock - 18 Coil||2/6||80||Diamond|
* Bench testing performed by an independent laboratory intended to evaluate time required to occlude 75% of blood flow. Time to reach 75% cessation of blood flow distal to coil placement, measured at 1-minute time increments. Note: 75% cessation endpoint used since can’t measure flow in a hollow tube – 25% flow was minimum amount of flow able to be measured in model
Data one file at BSC. n=7 in Concerto group. N=9 in Ruby group. Bench test results may not necessarily be indicative of clinical performance.
Interlock™ 35 and 18 and IDCTM Occlusion system is manufactured by Boston Scientific Corporation. All cited trademarks are the property of their respective owners. CAUTION: The law restricts these devices to sale by or on the order of a physician.
Indications, contraindications, warnings and instructions for use can be found in the product labeling supplied with each device. Information for the use only in countries with applicable health authority product registrations Not for use or distribution in France