InnovaTM is designed to allow precise and visible deployment with:
Proper deployment technique is critical to ensure predictable, controlled stent deployment - especially for 150, 180 and 200 mm stents.
1. Always begin deployment using the thumbwheel, and use the thumbwheel as long as possible
The thumbwheel alone should be used to deploy the first 120mm of stent length. Slow, controlled movements of the thumbwheel support even stent deployment and help control deployment forces.
2. For long-length stents (150+ mm), deployment must be completed by using the pull-grip
After 120mm of stent has been deployed using the thumbwheel, an arrow will become visible on the pull-grip handle, signaling that the pull-grip must be used to complete deployment. Slow, controlled movement of the pull-grip is essential to ensure proper deployment, especially for long stents.
Tray Label (150, 180 and 200mm stent trays)
Deployment arrow on pull-grip